5. Accessories Required.
The accessories required for
this calibration are common
usage accessories, issued as indicated in paragraph 4 above, and are not listed in this
Table 2. Minimum Specifications
of Equipment Required
Manufacturer and model
Minimum use specifications
Fluke, Model 5720A (5700A/EP)
(p/o MIS-35947) w/amplifier, Fluke,
Range: 1 to 100 V dc
Model 5725A/AR (5725A/AR)
Range: 1 to 1000 V ac
Range: -0.1 to 2.6 V dc
Fluke, Model 8840A/AF05
The instructions outlined
in paragraphs 6
and 7 are preparatory
to the calibration
process. Personnel should become familiar with the entire bulletin before beginning the
b. Items of equipment used in this procedure are referenced within the text by common
c. Unless otherwise specified, verify the result of each test and, whenever the test
requirement is not met, take corrective action before continuing with the calibration.
Adjustments required to calibrate the TI are included in this procedure. Additional
maintenance information is contained in TM 11-6625-2724-12 for this TI.
d. Unless otherwise specified, all controls and control settings refer to the TI.